Monitoring Parameters: SpO2, PR, PLETH, Pulse strength, TEMP
Signal sockets: SpO2 socket, PR socket, TEMP socket, USB socket(for transmit data and AC power)
2. Display screen
Type: 240 × 320 dot matrix color TFT LCD screen
Display area: 43mm×58mm
PLETH: The oximeter can display PLETH waveform
Audio and visual alarm: The speaker sounds the alarm and the parameters measurement will become in red to flash on the screen. Also the same when sensor-off.
Prompting: Icons of battery capacity, pulse sound switch and alarm sound switch.
Power saving: The oximeters will automatic shut down to save power. The time is settable.
Trend: Storage/Review trend data up to 3888 groups, data save time being set freely
Data storage: The saved data will never be lost even if the oximeter is powered off.
Communication: Transmitting the patients trend to a PC for display, observation, saving and printing via USB or Bluetooth communication
4. Power Requirements
Battery type: common 1.5V AA alkaline or rechargeable batteries
Run time: 6 hours
Operation: 0oC~ 45oC
Transportation and storage: -20oC ~ 60oC
Operation: 15% - 95 %( noncondensing)
Transportation and storage: 10% - 95 % (noncondensing)
Operation: 86 KPa ~106 KPa
Transportation and storage: 50KPa~106 KPa
6. Parameter Specifications
Patient: Adult, pediatric,neonate
Range: 0% - 100%
Resolution: 1 %
Accuracy: 70% - 99%: ?2 %
0% - 69%: Unspecified
Range: 0 bpm ~250bpm
Input: Body surface thermal-sensitive resistor temperature sensor
Range: 0oC ~50oC